For Immediate Release: Dec. 22, 2010Media Inquiries: Erica Jefferson, 301-796-4988, email@example.comConsumer Inquiries: 888-INFO-FDA
FDA announces recall of Abbott glucose test stripsStrips sold in retail stores, online and used in health care facilities
The U.S. Food and Drug Administration today announced the agency is working with Abbott Diabetes Care to recall 359 different lots of glucose test strips marketed under the following brand names:
Precision Xceed Pro;
ReliOn UltimaThese strips are used with Abbott’s Precision Xtra, Precision Xceed Pro, MediSense Optium, Optium, Optium EZ and ReliOn Ultima blood glucose monitoring systems. As many as 359 million strips may be affected by the recall. The blood glucose monitoring systems are not affected by this recall.
The recall pertains to certain lots of these test strips distributed in the United States and Puerto Rico. Other Abbott Diabetes Care products are not affected by the recall.
The test strips being recalled may give falsely low blood glucose results. False results may lead patients to try to raise their blood glucose unnecessarily, or they may fail to treat elevated blood glucose because of a false, low reading. Both scenarios pose risks to a patient’s health.
The recall is related to the test strips’ inability to absorb enough blood for monitoring. Strips exposed to warm weather or prolonged storage may be more likely to provide a false result.
The test strips were manufactured between January and September 2010 and are sold both in retail and online settings directly to consumers, but are also used in health care facilities.